System Function Test
After the spiral tube is loaded onto the enteroscope, a system check is performed to confirm correct loading and function.
Every time the control unit is turned on or the enteroscope has been replaced, the inspection mode is turned on. The spiral tube will always rotate counterclockwise first and then rotate clockwise in inspection mode.
The limit function is designed to protect the enteroscope’s rotation part and not intended to guarantee patient safety.
- Content Type