USG-400 Quick Reference Guide
This setup guide is a summary of the USG-400 Instructions for Use (IFU). For detailed operating instructions, be sure to follow the instruction manual that was included with your device when purchased.
SEAL & CUT Level
Delay activation time in between bipolar and ultrasound energy output. Bipolar comes first.
Minimum activation time of bipolar energy.
Intelligent Tissue Monitoring (ITM)
Immediate stop of bipolar and ultrasound energy supply after tissue transection.
1) The THUNDERBEAT System requires both the ESG-400 HF-generator and the USG-400 ultrasonic generator to be switched on.
2) In case an error occurs, please follow the instruction indicated on the display.
3) Cleaning of the instruments tip: Use a wet gauze or submerge it in a physiologic solution, activating the SEAL & CUT mode (purple button) with the branches in an open position.
- Content Type